Stainless steel laboratory fume cupboard for lab furniture equipment in college
Description: body is made of 1.0 mm stainless steel steel plate (304) in the numerical control processing center, clipping, positioning shape after punching, bending, welding, surface smooth without welds.
The inner description: 1.0 MM stainless steel plate. Equipped with removable arbors, facilitate maintenance circuit, water, gas path.
Fluorescent lamp: fluorescent lamp hidden on the panel, not contact with ventilation airflow inside ark, easy to replace. Using 30 w 1 fluorescent lamps, and is equipped with 5 mm toughened glass.
Stainless steel handle: inferior smooth handle.
Guide plate specification: 1.0 MM stainless steel plate.
Power supply: laboratory use special power socket.
Window: aluminum frame with plastic profile, window USES 3 block safety glass 5 mm, it can move around, and adopting balance device, it can move up and down, free to adjust.
Adjust a foot: the phi 10 mm diameter injection adjust a foot, shockproof, moisture, corrosion resistance, cabinet put oneself in another's position can be adjusted according to the indoor floor appropriate height, maximum adjustment of 0 to 30 mm.
The gas-collecting hood: use PP material
Distribution box: conform to the requirements of the 220 v and 380 v power supply, a leakage and motor protection device.
Control switch: 12 v touch switch, centralized control of the whole circuit system
Process description: all the steel plate welding after polishing processing, visual level without welds. All water, electricity, gas path requirements safe, applicable, and concealed installation. After the cabinet put oneself in another's back set maintenance hole.
Three pieces of guide plate space at different height of discharge harmful gases from different section area. Fume hood at the speed of surface wind speed 0.5 m/s operation will discharge air ventilation cabinet, ensure no any residual gas exists. Ventilation efficiency is high, the air volume is about 1600 m3 / h, and low noise.
The lower cabinet put oneself in another's position (plain bottom ark), door plank: use 1.0 mm stainless steel plate (double), no welding connection removable cushions. In the numerical control machining center, clipping, after the punching, bending, welding forming